|Place of Origin:||Ningbo,China|
|Minimum Order Quantity:||10 Piece/Pieces (Min. Order)|
|Price:||US $1-10 / Pieces|
|Packaging Details:||1 PCS /BOX|
|Delivery Time:||5 days|
|Payment Terms:||L/C, D/A, D/P, T/T, Western Union, MoneyGram|
|Supply Ability:||1000 Piece/Pieces per Day|
|Product Name:||Fiber Optic Electrode||Application:||For Fiber Optic|
|Color:||Black And Silver||Suitable:||FSU-10|
|Packing:||1 Pair /box||Certification:||ISO9001|
fiber splicing tools,
fiber optic cable tools
1.Life time for splice over 3000 times
2.Available for FSM-60S, FSM-60R, FSM-18S and FSM-18R Fusion Splicer
1.The ELCT series is the exclusive elctrodes for fusion splicers. In order to preform splicers properly and completely, the genuine electrodes shall be utilized. The ELCT series is supplued with a pair, for the anode and the cathode respectively.
Electrode cleaning and arc calibration:
1 After the electrode is installed, turn on the fusion splicer power switch.
2 After the fusion splicer enters the initial screen, press the menu button to enter the first level menu.
3 Find the machine maintenance by moving the cursor, enter the cleaning electrode item, after confirmation, the machine will automatically
Discharge clean and age the electrode.
4 Install the cut left and right fiber, find the arc position item by moving the cursor, and make the light through operation.
The fiber end face and the tip of the two electrodes maintain a three-point line position, and the machine will perform a discharge test.
5 Finally, press the menu button to launch
Do not touch the tip of the electrode during the replacement process to prevent damage and avoid the electrode falling inside the machine.
Calibration of the arc position after electrode replacement
1 First, you must confirm that the fusion splicer is in the off state, remove the protective cover of the electrode compartment, and loosen it.
Remove the upper electrode from the screw of the electrode; then loosen the top wire of the lower electrode and remove the lower electrode.
2 Secondly, install a new electrode. The operation steps are the opposite of the method of disassembling the electrode. want
Find the gap between the two electrode tips: 2.6 plus or minus 0.2 mm.
3 Finally, during the assembly process, the tip of the electrode should not be touched to prevent damage to the electrode and avoid the electrode
Dropped inside the machine and caused damage. After the electrode is replaced, the arc position should be calibrated.